CALIBRATION GAS GENERATORS
What is ACD Calibration Gas?
Simply put, Advanced Calibration Designs are gas calibration devices that are superior to the previous approaches to calibrating gas detection monitors by generating gas on demand as required.
Advanced Calibration Designs exceed the limitations of all previous forms of calibrating gas monitors. They do this through Integrated and Expandable products that provide the backbone of the new era in calibration gas standards.
- Most economical method to calibrate sensors and instruments
- Portable and easy to ship
- No inherent shelf life
- NIST traceable
- EASY to use
- No gas storage or wastage
- Expandable, one unit, many gases
- Gas supply or simple bump testing
Safe for use in hazardous environments and plane travel
Adjust concentration and flow rate on the fly
Many of products generate bout calibration gas and bump gas
Available with everything you need in a convenient calibration kit
ACD is proud to launch its NO2 cells for its long established CAL2000 ppm level calibration gas generator, expanding the range of gases that can be generated from a single hand held box even further.
This adds to H2S, Cl2, ClO2, H2 and NCN and is the most cost effective way to produce calibrated ppm level gas on demand. With extended shelf lives, NIST traceability and being easy to ship and transport the ACD generators are the way forward.
CHOOSE YOUR GAS
OR SIMPLY, CHOOSE YOUR CALIBRATOR
GAS PRODUCT CHART
|GENie QC - 1||GENie O3||GENie EC||CAL 2000||CAL 2000LT||CAL - 101||MiniBump|
|About This Product||Our most versatile calibration gas generator family.||A powerful calibration gas system offering adjustable or fixed concentrations.||A rugged calibration gas generating workhorse.||Calibration and bump test capabilities in a very small unit.|
|Adjustable||Source defined||Source defined||Source defined|
|Flow rate||Variable||0.2 @ 1.0 PPM Fixed flow capable||0.2 to 1 LPM Demand flow capable||0.2 to 1 LPM Demand flow capable||0.5 or 1 LPM||0.5 or 1.0 LPM||0.5 LPM|
|Accuracy||+/- 10%||+/- 5% or +/-10%||+/-5% or +/-10%||+/-5% or +/-10%||+/-5% or +/-10%||+/-20%||+/-20%|
|Ammonia||10 - 500 PPM|
|Chlorine||.05 - 50 PPM||.05 - 50 PPM||1 - 25 PPM||1 - 25 PPM||1 - 25 PPM|
|Chlorine Dioxide||.5 - 5 PPM||.5 - 5 PPM|
|Hydrogen||.5 - 500 PPM||.05 - 50 PPM||1 - 25 PPM||1 - 25 PPM||1 - 25 PPM|
|Hydrogen Cyanide||.05 - 50 PPM||.05 - 50 PPM||1 - 25 PPM||1 - 25 PPM||1 - 25 PPM|
|Hydrogen Sulfide||.05 - 50 PPM||.05 - 50 PPM||1 - 25 PPM||1 - 25 PPM||1 - 25 PPM|
|Nitrogen Dioxide||.05 - 5 PPM||.05 - 50 PPM|
|Ozone||.2 - 1 PPM|
|Electrochemical Ammonia||.05 - 100 PPM|